When I select tightness test on my KANE456/458, I get FULL on the display, why?
The memory section for tightness testing is full and the stored data needs to be deleted are full.
Turn the dial to Menu.
Scroll (use down arrow button) down to MENU REPORT and then press the enter button (Send button).
REPORT COMB’Nshould be displayed.
Scroll down to Report and press enter button.
Scroll down to Tightness and then press enter button
Scroll down to Delete All and press enter button.
Scroll down to No and press the enter button
Turn Dial out to Ratio
You should now be able to do the Tightness Test.