How do I pair a DTHA2 KANE LINK wireless anemometer to my KANE LINK FGA?
Rotate your FGA dial to LINK and select LINK using UP / DOWN then ENTER keys
Select ADD using UP / DOWN & ENTER keys then select DTHA2 using UP / DOWN then ENTER
Enter DTHA2 serial number using UP / DOWN then ENTER to enter 1 number at a time – ensure each serial number is exactly 10 digits.
If serial number is less than 10 digits, add zeros in front to reach 10 – for example, enter serial number 123456 as 0000123456
Please note you only do this once, even after replacing FGA batteries, until time to remove or replace your KANE LINK device
Switch on your DTHA2 & wait 10 secs – If paired, measurements appear on your FGA AUX screen
When finished, switch off your DTHA2 – Your AUX screen returns to its normal display
To reconnect to your KANE LINK FGA, switch on & wait 10 secs
Any questions, please call us on 08000590800