My CO2 reading is low?
Check that the water trap drain plug is fitted correctly in the water trap.
Check the hose of the probe for any signs of damage.
Check that the particle filter is dry and clean.
With the probe connected to the analyser, place the probe outside in fresh air, switch off then on again to set the zero correctly and repeat the analysis.