Why is KANE100 reading 0ppm CO2 in fresh air?
KANE100 needs to be set to ambient conditions.
This is done by using the true zero capsule (tube with green crystal). Connect the zero capsule to the KANE100 water trap and remove the end cap.
With the Kane100 switched on:
Turn dial to Menu
Use up button (print key) to Scroll to MENU ZERO and Press enter (hold) key.
Press up button to Zero True and press enter.
Analyser will now perform a Zero.
When finished remove Capsule form trap and turn dial to CO2.
You will now have a CO2 reading.